NEW WIRELESS HAND-HELD UNIT: THE PERIOTEST M
The Periotest M is the world's most commonly used, hand-held mobile device for measuring osseointegration in dental implants. Now it is even simpler to perform an objective evaluation of an implant's stability.
***This unit is Sold Out and will not be available until the beginning of the new year.
- Measures every type of implant on the market
- Provides An Objective Reproducible Measurement Confirming That The Implant Is Stable Enough To Load
- Detects Early Implant Failure, Measuring A Decrease In Stability As Subtle As 2-3%
- Measures Occlusal Load - And Overload
- Provides Objective Patient Record Documentation
- One-Button Operation
- Built-In Rechargeable Lithium Ion Battery Enables Permanent Readiness And Rapid Charging
Thanks to its wireless design, the user can now benefit from a maximum freedom of movement. It is no longer necessary to provide a surface for apparatus in the vicinity of the patient. The device's internal lithium-ion battery is simply recharged in the base unit provided, just like a cordless phone. What many implantologists are looking for is a simple and quick method of gaining an objective statement as to whether the implants they have made have become well osseointegrated and can be loaded. With the Periotest M, we have developed a device that is able to fulfil these requirements, and which can be seamlessly incorporated in the workflow of any dental practice.
The basic concept of the Periotest M is impressive because of its simplicity: the device is activated by a single switch and the measuring sequence begins forthwith. The test results are shown on the instrument's integral display. The new model retains the strengths of the traditional Periotest measuring procedure, particularly its suitability for use in universal applications: the instrument can be used for taking measurements on a wide variety of implants, without the need for any special accessories, such as a transmitter or similar device. The Periotest is clearly a very economical measuring instrument.
Moreover, the Periotest can be employed at all stages of implantological treatment – from primary stability testing, through the healing period, and right up to the finished prosthetic. It is this ability to measure the finished crown that is of particular value. According to reports, every tenth implant is susceptible to periimplantitis 15 years at the latest following its original insertion. By conducting continuous tests with the Periotest, it is possible to detect and treat this condition at an early stage.
It is therefore recommended that implants are measured routinely using the Periotest. Doing so will serve to build a sense of security and trust, both for the dentist and for the patient.
The Periotest M is a measuring device for use in dental practices and is designed for the following range of applications:
- Implants: Assessment of the osseointegration of dental implants
- Natural teeth: Diagnosis and assessment of periodontopathies,
assessment of the occlusal load and control of the treatment’s progress
The Periotest’s scale ranges from -8 to +50. The lower the Periotest value, the higher is the stability / damping effect of the test object (tooth or implant).
Assessment of the osseointegration of dental implants
Nowadays there are many implant systems which are in use. The Periotest M unit is able to take measurements on all of them. Moreover, measurements can be made at all stages of the implant process: directly after implantation, to measure primary stability, following the healing phase to determine that the required degree of osseointegration has taken place to enable pressure to be applied to the implant, and following completion of the prosthetic, to enable any negative developments to be recognised at an early stage.
As a result of the different implant systems and the existence of varying clinical conditions with individual patients, it is only possible to present guide values here. These show which Periotest value indicate good osseointegration and which are insufficient for withstanding pressure on the implant.
On the whole, implants lose a certain amount of stability in the first 14 days following implantation, and this is reflected by an increase in the Periotest values of one to two units. However, once the healing phase is over, the Periotest values obtained will be similar to those immediately following implantation (primary stability). Significantly greater increases in Periotest values, even a number of years after implantation, are an indication that the implant has become unstable, one of its screws has loosened, or it has excess pressure or has become infected (e.g. periimplantitis). It is therefore advisable to record all measurements to allow monitoring checks to be conducted over time.
Diagnosis and assessment of periodontopathies
While the Periotest value closely correlates with the tooth mobility, it is not the result of a conventional measurement of the tooth. The following correlations apply:
There are also tables for use with natural teeth, that indicate which Periotest values can be expected for a healthy periodontium. Please see the relevant literature for further information.
Early diagnosis of diseases of the periodontium:
Incipient or existing periodontal structural changes, including occlusal traumas, can be determined with considerable accuracy during routine examinations - before they can be detected on a dental roentgenogram.
Monitoring of treatment by follow-up examinations:
Objective, reproducible, and simple measurement permits the regular monitoring of the success of any periodontal, orthodontic, and functional therapeutic steps. The effect of occlusal disorders, and also adjustment, can be evaluated objectively. Monitoring of prophylactic measures, e.g. following removal of dental calculus, teaching oral hygiene, and periodontal therapy. Reduction of the radiation exposure associated with the monitoring of treatment.
Help in decision making for further treatment
Rapid and objective determination of the suitability of a treated tooth for use as an anchor for further therapeutic measures. Periodontal reconstruction and orthodontic procedures can be quantified.
Detection of occlusal overloading:
From the difference in the Periotest values during occlusion and in the absence of occlusion, occlusal overloading can be detected when this is greater than 6 Periotest values. This makes possible the immediate monitoring of occlusal adjustment in the case of almost all forms of occlusal treatments.
Periotest values are independent from such clinical parameters as condition of dental fillings, overlapping of teeth, presence of crowns or enamel defects.
HOW DOES THE PERIOTEST WORK
The Periotest measuring procedure is electromechanical. An electrically driven and electronically monitored tapping head percusses the test object (tooth or implant) 16 times. The entire measuring procedure requires approx. 4 seconds. The tapping head is pressure sensitive and records the duration of contact with the test object. Loose teeth or implants display a longer contact time and the Periotest values are correspondingly higher, while sturdy teeth or well osseointegrated implants have a short contact time and result in low Periotest values.
To ensure that the measurement is valid and meaningful, it is important to ensure that the handpiece is positioned correctly with respect to the test object (distance, angle). The device itself provides assistance by monitoring all 16 impulses. Incorrect impulses are eliminated to ensure reliable and reproducible measurements.